Post questions, comments, reviews or errors in the comment box below.
Your File is Ready … Download PDF
PDF Content Summary: Important: Remove this guide before handing the device to the patient. Pressures should only be adjusted by qualified personnel. S8 EliteTM II System Clinical Quick Setup Guide Setup is quick and easy ... Just Go! The following settings have been pre-set to accommodate typical patient treatment values. S8 Elite II Start CPAP: CPAP: Ramp: 4 10 20 cm cm minutes HH2O 2O Max Ramp: 30 minutes Humid: None SmartStart: On Reminders: Off Mask: Ultra Leak Alert: Off Tube Length: 2 m Language: English Smart Data: On Auto Appear: On Patient Menu: Std EPR: Ramp EPR Level: 1 EPR Inhale: Med EPR Access: On 1 1Accessing the clinical menu
• Turn on the device and wait until the standby (RAMP) screen appears on the LCD. At this stage you can adjust the ramp time.
• Press and simultaneously for three seconds to access the clinical menu. 2Accessing the quickview menu • Press and simultaneously for three seconds to access the quickview menu. 3Changing options and settings • Press to access standard machine settings, eg, Start CPAP pressure. • Use and to scroll through the RESULTS, OPTIONS and SERVICING menu screens. • From a menu screen, use to access the parameters in that menu. • Use and to scroll through menu parameters. • Use to view or change each parameter setting as required. • Press until you return to the standby (RAMP) screen. See the Clinical Guide for full details. 2
S8 Elite II clinical menus RAMP: 10 min menu CLINICAL MENU WELCOME CLINICAL menu exit SETTINGS enter exit RESULTS enter exit OPTIONS enter exit Settings Menu: screens Options Menu: screens SERVICING enter exit RUN HRS: 8888
exit SN:12345678912 exit PCB: 123456789123 4567 exit SW: SX4800100 exit BR: 19200 change exit 3 S8 Elite II clinical menus Settings Menu: screens SETTINGS enter exit MODE: CPAP change exit START CPAP: 4.0 change exit CPAP: 10.0 change exit MAX RAMP: 30 change exit MASK: ULTRA change exit TUBE LENGTH: 2m change exit HUMID: NONE change exit SMARTSTART: ON change exit LEAK ALERT: OFF change exit EPR: RAMP ONLY change exit EPR LEVEL: 1 change exit EPR INHALE: MED change exit EPR ACCESS: OFF change exit 4 S8 Elite II clinical menus Options Menu: screens OPTIONS enter exit SMART DATA enter exit REMINDERS MASK FIT: ON enter exit change exit FACTORY DEFAULTS reset exit ERASE DATA? yes exit DATE: 20 Dec 2006 change exit TIME: 15:54 change exit MENU TYPE: ADV change exit LANG: ENGLISH change exit Note: For each reminder, you can set a time period for that reminder to keep re-occurring. For these recurring reminders to work, you must set the date for the first occurrence of the reminder as well as a recurrence time period. For example, if you set MASK: 30Jun2008 and MASK RECUR: 6mth, the first reminder to replace the mask is displayed on 30 June 2008. The recurring reminder occurs 6 months from the date the reminder is acknowledged by the patient. The reminder will continue to recur at 6-month intervals unless it is set to OFF or the recurrence period is changed. MASK: 16Feb2007 PRESSURE: ON change exit change exit MASK RECUR: 6mth change exit USAGE: ON change exit CALL: 16Feb2007 AUTO APPEAR: ON change exit change exit CALL RECUR: OFF change exit CARD: 16Feb2007 change exit CARD RECUR: OFF change exit FILT: 16Feb2007 change exit FILT RECUR: 6mth change exit SERV: 16Feb2007 change exit SERV RECUR:24mth change exit CUST: OFF change exit CUST RECUR: OFF change exit 5 S8 Elite II quickview menu REVIEW SETTINGS enter exit enter exit MODE: CPAP exit START CPAP: 4.0 exit CPAP: 10.0 exit MAX RAMP: 30 exit MASK: ULTRA exit TUBE LENGTH: 2m exit HUMID: NONE exit SMARTSTART: ON exit LEAK ALERT: OFF exit EPR: RAMP ONLY exit EPR LEVEL: OFF exit EPR INHALE: MED exit RESULTS enter exit EFFICACY DATA enter exit LEAK: 0.70 L/s day... exit AHI: 4 day... exit AI: 3 day... exit ... ... ... ... ... USAGE DATA enter exit ... 6 EPR ACCESS: OFF exit PRESS: 10.2 PRESS: NO DATA day... exit yr... exit HI: 2 day... exit USED HRS: 0800 exit LEAK: NO DATA yr... exit AHI: NO DATA yr... exit AI: NO DATA yr... exit USAGE: 1000/1068 exit USAGE: 7.45 hrs days... exit HI: NO DATA yr... exit RUN HRS: 00000 exit USAGE: 7.2 hrs yr... exit S8 Elite II patient menus Detailed menus WELCOME >>>>>> RAMP: 10min menu CPAP SETTINGS enter exit EPR LEVEL: 1 change exit MASK: ULTRA change exit TUBE LENGTH: 2m change exit HUMID: NONE change exit SMARTSTART: ON change exit LEAK ALERT: OFF change exit RESULTS enter exit USED HRS: 08000 exit USAGE: 1000/1068 exit SMART DATA enter exit
OPTIONS enter exit SMART DATA enter exit LANG: ENGLISH change exit SERVICING enter exit AUTO APPEAR: OFF change cancel AUTO APPEAR: OFF change cancel
SN: 12345678912 exit PCB: 123456789123 4567 exit SW: SX4800100 exit This screen only appears if access is set by clinician. 7 S8 Elite II patient menus Standard menus WELCOME >>>>>> RAMP: 10min menu EPR LEVEL: 1 change exit USED HRS: 08000 exit USAGE: 1000/1068 exit MASK: ULTRA change exit SW: SX4800100 exit This screen only appears if access is set by clinician. ResMed Ltd (Manufacturer) 1 Elizabeth Macarthur Drive Bella Vista NSW 2153 Australia • ResMed Corp (US Designated Agent) 14040 Danielson Street Poway CA 92064-6857 USA • ResMed (UK) Ltd (EU Authorised Representative) 96 Milton Park Abingdon Oxfordshire OX14 4RY UK • ResMed Offices Australia, Austria, Belgium, Brazil, China, Finland, France, Germany, Greece, Hong Kong, India, Ireland, Italy, Japan, Malaysia, Mexico, Netherlands, New Zealand, Norway, Portugal, Singapore, South Africa, Spain, Sweden, Switzerland, UK, USA (see resmed.com for contact details). Protected by patents: AU 697652, EP 0661071, US 5199424, US 5522382, US 6240921, US 6705315. Other patents pending. Protected by design registrations: AU 302180, AU 302181, AU 302182, AU 302183, AU 302184, AU 302185, AU 302186, CN 200430121212.2, CN 200430121213.7, CN 200430121211.8, CN 200430121214.1, CN 200430121215.6, CN 200430121216.0, CN 200430121217.5, EU 269436, EU 815071, HK 0412901.9, JP 1257662, JP 1257663, JP 1266956, JP 1266957, JP 1267270, JP 1267271, NZ 405614, NZ 406923, NZ 406924, NZ 406925, NZ 406926, NZ 406927, NZ 406928, US D544598, US D553228, US D557406, US D557407, US D560795, US D561891, US D565160. Others pending. S8, S8 Elite and SmartStart are trademarks of ResMed Ltd. ©2008 ResMed Ltd. 338619/1 08 07 8 Global leaders in sleep and respiratory medicine resmed.com
FREE ENGLISH PDF
USER GUIDE - USER MANUAL
OWNER GUIDE - OWNER MANUAL
REFERENCE GUIDE - REFERENCE MANUAL
INSTRUCTION GUIDE - INSTRUCTION MANUAL