Post questions, comments, reviews or errors in the comment box below.
Your File is Ready … Download PDF
PDF Content Summary: 4.5” 4.5” 4.5” Setup Overview Refer to your User’s Manual for full System instructions and information. Reader Touchscreen Home Button ● 1 Assemble and Apply Sensor to the back of your upper arm Place Sensor Applicator over site and push down firmly to apply Sensor. CAUTION: Do not push down on Sensor Applicator until placed over prepared site to prevent unintended results or injury. USB Port ● 1 Assemble and Apply Sensor to your body FLASH GLUCOSE MONITORING SYSTEM Quick Start Guide for Sensor ● 2 Start new Sensor with Reader Applicator FreeStyle Libre 2 Reader Sensor Pack Codes on labels ● 3 Check your glucose whenever you want for up to 14 days must match STEP 1 STEP 4 STEP 7 Wait 60 minutes for start-up Select site on back of upper arm. Do not use other sites as these may result in inaccurate glucose readings. Note: Avoid scars, moles, stretch marks, lumps, and insulin injection sites. To prevent skin irritation, rotate sites between applications. Lid Cap Line up dark mark on Sensor Applicator with dark mark on Sensor Pack. On a hard surface, press down firmly on Sensor Applicator until it comes to a stop. Sensor Gently pull Sensor Applicator away from your body. Make sure Sensor is secure. Discard used Sensor Applicator and Sensor Pack according to local regulations. Wash site using a plain soap, dry, and then clean with alcohol wipe. Allow site to air dry before proceeding. Lift Sensor Applicator out of Sensor Pack. STEP 2 STEP 5 STEP 8 CODE XXX Peel lid completely off Sensor Pack. Unscrew cap from Sensor Applicator. CAUTION: Sensor codes must match on Sensor Pack and Sensor Applicator or glucose readings will be incorrect. Sensor Applicator is ready to apply Sensor. CAUTION: Sensor Applicator now contains a needle. Do not touch inside Sensor Applicator or put it back into Sensor Pack. STEP 3 STEP 6 STEP 9 FreeStyle Libre Sensor Pack | Envase del sensor Customer Service | Atención al cliente: 1-855-632-8658 CODE XXXXXXXXXXX Abbott D XXX YYYY-MM-DD 4.5” 4.5” 4.5” ● 2 Start new Sensor with Reader STEP 2 ● 3 Check your glucose STEP 2 STEP 3 STEP 1 STEP 2 Using Alarms Touch Dismiss Alarm & Check Glucose or press the Home Button to dismiss alarm and check your glucose. Important OK Low Glucose Alarm Dismiss Alarm & Check Glucose OK Do not take high doses of vitamin C (more than 500 mg per day). This may falsely raise your Sensor readings. Supplements like Airborne® or Emergen-C® have high doses of vitamin C. Read labeling for all supplements to determine vitamin C content. Press Home Button to turn on Reader. Touch Start New Sensor. Note: If using Reader for the first time, follow the prompts to set date Hold Reader within 1.5 inches (4 cm) of Sensor to scan it. and time. Press Home Button to turn on Reader. Setting Alarms Settings Alarms Sound & Vibration Time & Date 1 / 3 Press Home Button to go to Touch Alarms and then touch Home Screen. Touch . Change Alarm Settings. High Glucose Alarm Signal Loss Alarm Scan Sensor now? Dismiss Alarm & Check Glucose No Yes Touch No to dismiss alarm. Touch Yes or press the Home Button to dismiss alarm and scan the Sensor. If you have followed the instructions described in the User’s Manual and are still having difficulty setting up your System or if you are not sure about a message or reading, contact your health care professional. 10:23pm Check Glucose Ends in 14 days STEP 1 STEP 1 Review History Message Touch for more information Current Glucose Glucose from your latest scan Glucose Graph Graph of your current and stored glucose readings The Sensor automatically communicates with the Reader and can give you glucose alarms. Alarms are on by default. To change their settings or turn alarms off, follow these steps. IMPORTANT: Glucose alarms are an important safety feature. Please consult your health care professional before making changes. You must scan your Sensor to see your glucose reading. The shape of the circle sensor unit, FreeStyle, Libre, and related brand marks are owned by Abbott. ©2019 Abbott ART40392-001 Rev. C 06/20 Review the important information on the screen. Sensor can be used to check your glucose after 60 minutes. Hold Reader within 1.5 inches (4 cm) of Sensor to scan it. Reader shows your glucose reading. This includes your Current Glucose, Glucose Trend Arrow, and Glucose Graph. Select and set your alarms. Touch done to save. 10:23pm No Active Sensor Start New Sensor Review History OK Alarm Settings OK Change Alarm Settings Low Glucose 70 mg/dL Low Glucose Alarm High Glucose 240 mg/dL High Glucose Alarm Signal Loss On Signal Loss Alarm Change Alarm Settings When any alarm is on, sound/vibration and signal status display on the Home Screen. Symbol What it means 10:23pm Sound and Vibration ON Ends in 14 days Sound and Vibration OFF Sensor communicating with Reader Check Review Sensor not communicating with Reader Glucose History Refer to User's Manual for additional symbols. Notifies you when glucose is above the level you set. STEP 3 Glucose Going Low 82 mgdL 350 250 15050 2pm 6pm 10pm STEP 3 Manufacturer Consult instructions for use Glucose Trend Arrow What it means Glucose rising quickly Glucose rising Glucose changing slowly Glucose falling Glucose falling quickly New Sensor Starting UpSensor can be used in: 60 Notifies you when glucose is below the level you set. Notifies you when Sensor is not communicating with Reader and that you are not getting Low or High Glucose Alarms. minutes
FREE ENGLISH PDF
USER GUIDE - USER MANUAL
OWNER GUIDE - OWNER MANUAL
REFERENCE GUIDE - REFERENCE MANUAL
INSTRUCTION GUIDE - INSTRUCTION MANUAL